Simple automatic calibration method
Press A/M key while holding down ▼ key in the automatic state, then release at the same time, then the automatic calibration is started (the same function as the method 2 automatic calibration) .
Error Code List
|Control signal interrupted or less than 0.3 mA
|Signal feedback line or open-close line connection between positioner and actuator or program operation error
|The large oscillation of actuator may be due to unstable input signal or feedback signal, high precision, etc.
|Stop turning when the actuator moves to the closed direction
|The actuator stops when it goes in the open direction
|The temperature in the housing of the positioner exceeds 70 °C